2.first rinse the electrode with distilled water, and suck it with filter paper.
3.turn the meter on by pressing the key.
4.immerse the ph meter electrode in the solution to be tested(can not be over the immersion line )
5.stir gently and wait around 30 seconds till the reading stabilized.
6.after finished ,clear the electrode with pure water ,turn the meter off by pressing the ON/OFF key.
7.Always replace the protective cap after use.
The instrument need recalibrate in the following condition.
1.Long time no use or use very long time after calibration
2.The electrode is used very frequently
3.The testing accuracy requirement is very high
4.Press the "CAL"button long time in the air carelessly
Always replace protective cap after using digital meter to keep electrode from drying out due to prolonged exposure to air,which leads to slow and/or unstable readings.
If electrode has been dried out immerse it into distilled water for a few hours.
When the display value is fuzzy or unshown , The battery should be replaced promptly.Pay attention to the polarity of battery.
The instrument is warranted to the purchaser for a year from the date of purchase.
If during these periods, repair or replacement of the instrument is required,please return the instrument to either your dealer or to our office.The repair will be effected free of charge. ( damage is not due to negligence or erroneous operation by the user)
Note :Plese provide purchase invioce or proof before returning your instruments back .